Professional Reference articles are designed for health professionals to use. They are written by UK doctors and based on research evidence, UK and European Guidelines. You may find one of our health articles more useful.
Synonym: AV dyssynchrony syndrome
This article deals with the complications of pacemakers, including pacemaker syndrome. Pacemakers are discussed further in the separate articles Pacemakers and Pacing, Inserting Temporary Pacemakers, and Implantable Cardioverter Defibrillators.
- Pacemakers provide electrical stimuli to cause cardiac contraction during periods when intrinsic cardiac electrical activity is inappropriately slow or absent.
- Pacing systems consist of a pulse generator and pacing leads.
- Pacemaker output generally stimulates the cavity of the right atrium and/or right ventricle (endocardial pacing). Alternatively, epicardial leads can be implanted surgically on to the heart's surface.
- The rate of acute complications of pacemaker insertion is 4-5% and mostly related to operator experience.
- The incidence of late complications of permanent pacemakers has been reported as 2.7%.
- Failure to output: no pacing spike is present despite an indication to pace. This may be due to battery failure, lead fracture, a break in lead insulation, oversensing (inhibiting pacer output), poor lead connection at the take off from the pacer, and 'cross-talk' (ie a phenomenon seen when atrial output is sensed by a ventricular lead in a dual-chamber pacer).
- Failure to capture: pacing spike is not followed by either an atrial or a ventricular complex. This may be due to lead fracture, lead dislodgement, a break in lead insulation, an elevated pacing threshold, myocardial infarction at the lead tip, certain drugs (eg, flecainide), metabolic abnormalities (eg, hyperkalaemia, acidosis, alkalosis), cardiac perforation, poor lead connection at the take off from the generator, and improper amplitude or pulse width settings.
- Oversensing: pacer incorrectly senses electrical activity and is inhibited from correctly pacing. This may be due to muscular activity, particularly oversensing of the diaphragm or pectoralis muscles, electromagnetic interference, or lead insulation breakage.
- Undersensing: pacer incorrectly misses intrinsic depolarisation and paces despite intrinsic activity. This may be due to poor lead positioning, lead dislodgment, magnet application, low battery states, or myocardial infarction.
- Pacemaker tachycardia.
- Pacemaker syndrome: some patients with VVI pacemakers, especially with sinoatrial (SA) rather than atrioventricular (AV) disease, will show retrograde ventriculoatrial (VA) conduction during ventricular pacing which can cause fatigue, dizziness and hypotension. Pacemaker syndrome is associated with atrial cannon waves caused by simultaneous atrial and ventricular contractions. Replacement with a dual chamber is required.
- Operative failures:
- Pneumothorax (may require chest drain).
- Air embolism.
- Erosion of the pacer through the skin (rare - requires pacer replacement and systemic antibiotics).
- Haematomas (may require drainage).
- Lead dislodgement - usually occurs within two days following implantation of a permanent pacer and may be seen on chest radiography (if the lead is floating freely in the ventricle, malignant arrhythmias may develop).
- Venous thrombosis - rare and usually presents as unilateral arm oedema.
Complications of temporary pacing
- Immediate complications include:
- Late complications include:
- Ventricular arrhythmias.
- Septicaemia (especially staphylococcal infection).
- Wrong position requiring repositioning.
- Blood tests: electrolytes, coagulation screen, digoxin levels if appropriate, myocardial injury markers - eg, troponins (may indicate recent myocardial infarction)
- 12-lead ECG - any sign of myocardial ischaemia, arrhythmias or abnormal sensing.
- CXR: evaluate lead position and look for lead fracture. A CXR can be used to identify the pacemaker model, as most pacemakers have an X-ray code which is visible on a standard CXR.
- Echocardiogram: to assess for lead position, pericardial effusion or tamponade, or lead fracture.
- Pacemaker assessment:
- Review triggering thresholds, lead impedance, and battery function. Some brands may have stored electrocardiograms.
- Magnet application:
- This will result in asynchronous pacing mode and can reveal loss of capture and battery depletion.
- Telemetry monitoring:
- Usually useful in the early post-implant period; can reveal loss of capture.
- Transtelephonic monitoring:
- Rarely used and is most helpful in determining battery depletion.
- To evaluate lead fracture.
Pacemaker syndrome refers to the occurrence of symptoms relating to the loss of AV synchrony in patients with a pacemaker, ie the atria contract at the same time as paced ventricular contractions.The result is decreased cardiac output in response to which the total peripheral resistance will increase. But not all patients exhibit a rise in blood pressure and indeed some will have a drop in their blood pressure.
In 1994 Furman redefined pacemaker syndrome as:
- Loss of AV synchrony.
- Retrograde VA conduction.
- Absence of rate response to physiological need.
The incidence of pacemaker syndrome has been estimated to range from 7% (symptoms severe enough to warrant pacemaker revision) to over 80% (mild to moderately severe symptoms).Asymptomatic pacemaker syndrome is probably common and the true incidence of pacemaker syndrome much higher.
- Sick sinus syndrome as have preserved AV conduction.
- Single-chamber ventricular pacing.
Presentation of pacemaker syndrome
- Symptoms can vary considerably and also vary in severity.
- Symptoms include pulsation and fullness in the neck, dizziness, palpitations, fatigue, light-headedness and syncope.
- Symptoms and signs of heart failure may occur.
- Signs include hypotension, tachycardia, tachypnoea, raised JVP and cannon waves.
- There may be variations in pulses and fluctuating blood pressure - a drop of 20 mm Hg or more during ventricular pacing compared with that during atrial or AV synchronous pacing is suggestive.
- Pseudopacemaker syndrome - AV dyssynchrony may also occur without a pacemaker - eg, severe first-degree AV block, and hypertrophic cardiomyopathy with complete AV block.
- Other causes to consider - pacemaker malfunction, worsening heart failure, iatrogenic, and autonomic dysfunction.
- In patients with other pacing modes, symptoms usually resolve after upgrading the pacemaker to a dual-chamber pacing system, or reprogramming the pacemaker parameters - eg, AV delay, post-ventricular atrial refractory period, sensing level, and pacing threshold voltage.
- Medical therapy has a limited role but electrolyte abnormalities may need to be corrected and the medication regime should be reviewed and adjusted as needed.
- Use atrial pacing rather than ventricular.
- Use of dual-chamber pacing with long AV interval (provided marked first-degree AV block is not present).
- Control AV delay, to achieve physiological timing of atrial and ventricular contractions.
- Once the pacemaker has been optimised as above the symptoms resolve fully.
Further reading and references
; New pacing technologies for heart failure. BMJ. 2003 May 17326(7398):1073-7.
; Cardiac pacing: the state of the art. Lancet. 2004 Nov 6-12364(9446):1701-19.
; American College of Cardiology/American Heart Association/North American Society for Pacing and Electrophysiology, 2002
; Current practice and complications of temporary transvenous cardiac pacing. BMJ. 1996 May 4312(7039):1134.
; Diagnosis of pacemaker syndrome by suprahepatic vein pulsed Doppler echocardiography. Eur J Echocardiogr. 2011 Mar12(3):E25. doi: 10.1093/ejechocard/jeq186. Epub 2011 Jan 7.
; Pacemaker syndrome. Pacing Clin Electrophysiol. 1994 Jan17(1):1-5.
; New concepts in pacemaker syndrome. Indian Pacing Electrophysiol J. 2004 Oct 14(4):195-200.
; Pacemaker syndrome and pseudo-ventricular high threshold after dual-chamber pacemaker replacement. Europace. 2010 Dec12(12):1795-6. doi: 10.1093/europace/euq265. Epub 2010 Jul 16.
; NICE Technology Appraisal Guidance, February 2005
Hello . I apology for disturbing you and thanks for you time. From almost 3 year if i walk quick or lift any weight and then rest. My heart sinking and feels its muscels taking much time to become...Fearheartproble
Disclaimer: This article is for information only and should not be used for the diagnosis or treatment of medical conditions. Patient Platform Limited has used all reasonable care in compiling the information but make no warranty as to its accuracy. Consult a doctor or other health care professional for diagnosis and treatment of medical conditions. For details see our conditions.